SeQual Eclipse 3 Portable Concentrator - SKU : 5900
SeQual Eclipse 3 Portable Concentrator Product Overview Back to top
Eclipse 3 can operate in both Pulse Dose and Continuous Flow Modes. It operates in Pulse Dose or Continuous Flow Modes. In Pulse Dose Mode, oxygen is supplied in a burst at the beginning of each inspiration, providing a selectable range setting of 16 to 192 mL. In Continuous Flow Mode, the oxygen is provided at a constant flow rate between 0.5 and 3.0 liters per minute.
The battery life can last for up to 3 hours (depending on setting) and can be charged while in use. The unit can operate on either internal rechargeable battery power or an external power source. It includes a Smart Battery charger that recharges the internal battery whenever the unit is connected to any external power source.
It features an easy to transport cart with an effortless telescoping handle to transport the unit over even the toughest terrain. This is one of the most user-friendly portable concentrators available today. It also comes bundled with a variety of accessories to help support the active lifestyle, including a accessory bag and cart.
The SeQual Eclipse is approved by the FAA (Federal Aviation Administration) for use during flights on most major airlines. You can also use it on buses, trains, cruise ships and boats.
Don’t be tied down to oxygen tanks or large home concentrators. Get your freedom back!
The SeQual Eclipse Starter Package includes the Eclipse unit, one rechargeable battery, AC and DC adapters, cart, accessory bag, cannula and manual. Please view our Eclipse 5 Starter Package.
Product Specifications Back to top
|Product Features||SeQual’s Patented autoSAT Technology.
Pulse Dose Mode will deliver a continuous flow of oxygen, always searching for a breath about every 15 seconds.
Approved for Prescription Oxygen in continuous flow mode of 0.5 - 3.0 liters per minute.
LCD screen that shows LPM and mL on the display, as well as easy touch control settings that function on fast, medium and slow settings.
The Universal Cart has a new modern design that allows for trouble-free access to battery. Both AC and DC power adapters are smaller and lighter weight, yet more durable. The AC/DC and battery power choices increase therapy options and allow for more travel opportunities.
FAA approved for most commercial flights. Also approved for use on buses, cruise ships and most other methods of travel.
Wide array of accessories to help support an active lifestyle
Easy access to maintenance and service information
|Dimensions||12.3 inches Wide x 7.1 inches Deep x 19.3 inches High|
|Product Weight||15 pounds, 18.4 pounds with battery|
|Warranty||Three-year Manufacturer’s Warranty on Unit
One-year Warranty on Accessories
|Alarms / Alerts||Pulse Dose Mode, Lower Battery Power, Loss or No Power, Unit Failure, No Inspiration Detected, Low Therapeutic O2 Output, O2 Flow Outside Normal Limits|
|Battery Charging Time||It may take up to 5 hours, depending on the flow setting, to achieve 80 percent capacity from a completely uncharged battery.|
|Battery Duration||Up to 3.7 hours at 1.0 LPM continuous flow
Up to 1.3 hours at 3.0 LPM continuous flow
Up to 4.4 hours at 1.0 pulse flow setting
Up to 2.1 hours at 6.0 pulse flow setting
|Flow Settings||Pulse Dose: 16 - 96 in 16 mL increments, 128, 160 and 192 mL
Continuous Flow: 0.5 to 3.0 Liters Per Minutes in 0.5 increments
|Maximum Oxygen Output||3000 ml/min|
|Noise Level||40 Decibels @ 3.0 Pulse Dose, 48 Decibels @ 3 Liters per minute on continuous flow|
|Operating Ranges, Intended for Use||Altitude: Sea level to 13,123 feet
Temperature: 50 to 104 degree Fahrenheit @ 82.4 Fahrenheit non-condensing, dew point.
|Outlet Pressure||5.0 Nominal PSI (Pounds per Square Inch)|
|Oxygen Concentration||90% @ sea level (+/- 3%)|
|Power Consumption||52 watts at 1.0 and 145 watts at 3.0 Continuous Flow
45 watts at 1.0 and 95 watts at 6.0 Pulse Dose
|Trigger Sensitivity||Adjustable trigger ranging from -.125 cm to -0.4 cm|
|Power||AC Operation: 100-240 VAC, 50-60 Hz
DC Input: 13.5-15 VDC, 10A
Summary of Customer Ratings & Reviews Back to top
Review by Bob
Review by June
Review by Steve Shaver
Review by Tom Douthitt
Review by Catherine
Review by Loraine Dressler
Review by Robert Cue
Review by Greg C
Review by Fayza A.
Review by Alphonse W.
Product Q & AHow to change filter on the SeQual Eclipse 3?
The SeQual Eclipse 3 has a washable/external filter and an internal filter that need to be maintained. The washable/external filter is located on the back side of the machine just below the carry handle and is a mesh square filter. The gross particle filter should be rinsed off and completely dry before placing back into the machine. Depending on the number of hours the machine is used each day, the internal filter should be replaced every 1-2 years.
How do I perform a hard reset on the SeQual eclipse 3?
To perform a hard reset, you will need to first remove the battery from the machine and disconnect all power to the unit. Let the machine sit powered down for 2 minutes. Plug the machine back into AC power (the power supply for use in the home) WITHOUT BATTERIES and turn the machine on. Once the unit has been running for 5-10 minutes, place the battery into the machine WHILE RUNNING. This should eliminate the issue the machine is having. If the issue still occurs, call us and we will begin the process of creating a case.
How do I insert and remove the battery on the SeQual Eclipse 3?
The easiest way to insert and re