Complete the form below.
Respironics EverGo Portable Concentrator - SKU : 900-000
Respironics EverGo Portable Concentrator Description Back to top
It gives you the freedom to travel anywhere you can imagine. Because it weighs less than 10 pounds and comes with a carry bag and cart for easy transportation, it is ideal for people who are active and love to travel. The EverGo is FAA approved for use on airplanes, and it is designed to fit flawlessly on all commercial aircraft.
The EverGo unites pulse dose technology with 1050 ml/min of oxygen capacity to ensure you obtain the perfect amount of oxygen. It detects inspiration on the onset and delivers a measured dose of oxygen right when it is needed. The pulse setting of the EverGo creates the equivalent in liters per minute (LPM) of oxygen from a continuous flow machine and delivers the oxygen only when you breathe to avoid any waste.
The LCD touch screen features controls that are easy to read and enable you to adjust the settings on the EverGo. The unit is low maintenance with a washable particle filter that is easily accessible for cleaning.
The Respironics EverGo is DISCONTINUED, please consider the Respironics SimplyGo or call one of our agents to discuss our used inventory.
Product Specifications Back to top
|Product Features||Includes 2 rechargeable batteries that can give you up to 12 hours of operation life that are easy to charge, remove and change. The batteries can be charged with any outlet or vehicle charger. It can be used while the unit is charging.
User friendly LCD touch screen that controls all functions and displays battery life and your prescribed settings.
Features an innovative design that weighs less than 10 pounds and looks like a camcorder case.
Low maintenance with one filter that is easy to remove and clean.
Approved for airline travel by the Federal Aviation Administration (FAA)
|Dimensions||6 inches Wide x 12 inches Deep x 8.5 inches High|
|Product Weight||8.5 pounds with a single battery, 10 pounds with 2 batteries|
|Warranty||Three-year Manufacturer’s Warranty
90-Day Manufacturer’s Warranty on Accessories
|Battery Charging Time||Approximately 2 to 3 hours for each battery from 0% charge to 100%. Various factors determine charge times; usage and the power source can extend the time it takes to completely recharge batteries.|
|Battery Duration||On a setting of 2 at 20 Breaths Per Minute
1 Battery: Up to 4 hours
2 Batteries: Up to 8 hours
|Flow Settings||Pulse Dose: 1 to 6 in 0.5 increments|
|Noise Level||Around 43 decibels (As loud as a refrigerator humming)|
|Operating Ranges, Intended for Use||Temperature: 41 - 104 Fahrenheit (5 - 40 Celsius)
Humidity: 15% - 95%
Altitude: 0 - 8,000 feet (0 - 2438 meters)
|Outlet Pressure||3 PSI (Pounds Per Square Inch)|
|Oxygen Concentration||All Settings: 90% (+/-3%)|
|Storage Ranges, Intended for Storage||Temperature: 41 - 104 Fahrenheit (5 - 40 Celsius), unit only
Humidity: 0% - 95%, non-condensing
Store in a Dry Environment
|User Interface||LCD TouchScreen|
Respironics EverGo Portable Concentrator ReviewsBack to top
Review by canyondweller
Review by jerry p
Review by Mary Ellen DiPasquale
Review by Cathy Dumolien
Review by LC
Review by Richard Medlin
Review by Joe V.
Review by Jaco T.
Review by Debbie J.
Review by Sonya L.
Product Q & AHow to change filter on the Respironics Ever Go?
The EverGo has a washable/external filter and an internal filter that need to be maintained. The washable/external filter is located on the side of the machine. It is a small rectangular cutout covered by mesh netting and has a zipper. The gross particle filter should be rinsed off and completely dry before placing back into the machine. Depending on the number of hours the machine is used each day, the internal filter should be replaced every 1-2 years.
How do I perform a hard reset on the EverGo?
To perform a hard reset, you will need to first remove both batteries from the machine and disconnect all power to the unit. Let the machine sit powered down for 2 minutes. Plug the machine back into AC power (the power supply for use in the home) WITHOUT BATTERIES and turn the machine on. Once the unit has been running for 5-10 minutes, place the batteries into the machine WHILE RUNNING. This should eliminate the issue the machine is having.If the issue still occurs, call us and we will begin the process of creating a case.
How do I insert and remove the battery on the EverGo?
When looking at the unit from the top, you will see a Velcro piece that is fastened to the machine. Fold the Velcro piece up and you will notice a zipper. The top half of the bag on the machine will unzip, sort of like a lunch box. This is where the batteries are located. When placing the battery inside the machine, make sure to set it in gently. If you drop or slam the battery into the machine, the tabs, pc boards on the battery or the inside of the machine can get damaged or break. When removing the battery, place your finger on the plastic release and either push or pull away or toward you, then lift up on the black tab on the battery itself. Lift up on the battery and remove toward you, making sure you have a firm grip on the battery so it does not drop back into the machine or slip out of your hand.
How long does the battery last at all settings on the EverGo?
The battery duration FULLY CHARGED for Pulse Only up to the setting of 6. Keep in mind that battery durations are estimated because each patient has different breath rates and breathing patterns, which will make the batteries last for shorter or longer amounts of time.
With 1 battery in the machine
Setting 1: 6 Hours
Setting 2: 4 Hours
Setting 3: 3 Hours
Setting 4: 2 Hours
Setting 5: 2 Hours
Setting 6: 2 Hours
With 2 batteries in the machine
Setting 1: 12 Hours
Setting 2: 8 Hours
Setting 3: 6 Hours
Setting 4: 4 Hours
Setting 5: 4 Hours
Setting 6: 4 Hours