A+ BBB Rating
Excellent Trustpilot Customer Rating
Price Match Guarantee
- Two Columns Included
- No Tools Required
Inogen One G3 Replacement Columns (Flow Setting 1-4) Description
Two columns are included with each order.
NOTE: The RP-320 replacement columns are ONLY compatible with the G3 with that has FLOW SETTINGS of 1-4. If your flow setting is adjustable to Setting 4, this option is for you.
A low oxygen purity alert on the LDC screen will display “O2 Service Soon” when the columns need to be changed. To remove the columns, simply press the latch release button located on the bottom of the unit then slide the columns (metal tubes) out of the unit. No special tools are required!
View the Inogen One G3 1-5 Replacement Columns or the Inogen One G3 1-5 Replacement Columns.
Two columns are included
No tools necessary
|Dimensions||2" Wide x 2" Long x 5" High|
|Product Weight||10 ounces (each column)|
Learn how to replace columns for the Inogen One G3 portable oxygen concentrator. This video instructs you on how to remove the columns, install new columns, prepare for reset, and to perform a sieve reset. G3 Column Replacement Step...