autoSAT Technology will maintain a steady level of oxygen with Adjustable Rise Time and Bolus Size
Continuous flow from .5 to 3 LPM, Pulse Dose volume ranges from 16 to 192 mL
Battery recharges while the unit is connected to any heavy-duty DC power supply
All flow settings (continuous and pulse) are available while the machine is operating on DC power
Upgraded internal system will increase unit dependability
Three rise time options: slow, medium and fast with three adjustable pulse dose sensitivity settings
Easy access to removable battery with new contemporary mobile cart design
12.3 inches wide x 7.1 inches deep x 19.3 inches high
15 pounds (unit only), 18.4 pounds with battery
Three-Year Manufacturer’s Warranty on Unit
One-Year Warranty on Accessories
Alert / Alarm:
Unit malfunction, Low battery, Loss of unit power, Oxygen flow outside normal limits, Low therapeutic oxygen output, No inspiration detected in pulse dose mode
Back Up Alarm Power:
9 volt battery
1.8 to 5 hours to reach 80% battery capacity (dependent upon flow level)
Continuous Flow: 2 hours at 2 LPM
Pulse Dose Mode: 5.1 hours at a setting of 2
Pulse Dose Settings: 16 to 96 mL, 128 mL, 160 mL, 192, mL
Continuous Flow Settings: 0.5 liter increments from 0 .5 to 3 (LPM)
Green Light: Normal, Yellow: Caution, less than 85% power, Red Flashing: warning, less than 70% power
48 decibels at 3 LPM on continuous flow (Average office noise)
40 decibels at a 3.0 pulse dose setting (Quiet conversation)
Altitude Range: Sea level to 13,123
Temperature Range: 50 degrees to 104 degrees F at 82 degrees F dew point, non-condensing
87% to 95% at sea level
AC Power: 100-240 VAC, 50-60 Hz
DC Power: 12 V nominal
Battery: 97.5 W-Hr battery packs each containing 7.92 gm of lithium content
Continuous = 52 watts at 1 LPM - 145 watts at 3 LPM
Pulse Dose = 45 watts at 1, 95 watts at 6
Negative pressure between -.135 to -.37 cm
Is Sequal Eclipse 5 Approved by FAA?
Yes, the SeQual Eclipse 5 is approved by the FAA and meets all FAA requirements.
DC Power Information
It is important to understand that even though 3 LPM and battery charging are available on the Eclipse 5; the car’s electrical system will be the determining factor as to whether or not these features will be available. If the car’s outlet is not providing sufficient continuous power, the Eclipse 5 may not run properly.
Helpful Hints to Improve Functionality
Always have your car operating before you plug the Eclipse 5 into the DC outlet.
Use the DC accessory outlet closest to the car’s battery. This one tends to have the least power fluctuations and draws the most power.
Test the unit in your vehicle before a road trip to ensure proper functionality.
Start the Eclipse at the lowest continuous flow setting, 0.5 LPM
Increase the continuous liter flow incrementally, testing each setting to ensure that the Eclipse does not alarm
*This product is available for FREE GROUND SHIPPING. Please mention code FREESHIP2015 to the sales representative or enter it under "Discount Codes" in your shopping cart.
Please allow for 3-6 business days for UPS Ground Delivery in the US: most orders are processed within 1 business day. This item can be expedited for an additional fee. Additional charges apply to products shipped to AK, HI and territories outside the continental US. Free shipping is only eligible on new portable concentrators and does not apply to AK or HI. Cannot be combined with other offers. Restrictions may apply.
Other Delivery Options:
UPS 2nd Day: Delivers the second business day.
UPS Next Day: Delivers the next business day.