SeQual’s Patented autoSAT Technology.
Pulse Dose Mode will deliver a continuous flow of oxygen; always searching for a breath about every 15 seconds.
Approved for Prescription Oxygen in continuous flow mode of 0.5 - 3.0 liters per minute.
LCD Screen that shows LPM and mL on the display as well as easy touch control settings that are available in in Fast, Medium and Slow.
The Universal Cart has a new modern design that allows for trouble-free access to battery. Both AC and DC power adapters are smaller and light weight yet are more durable. The AC/DC and battery power choices will increase therapy and travel time.
FAA Approved for most commercial flights. Also approved for use on buses, cruise ships and most other methods of travel.
Wide array of accessories to help support the active lifestyle
Easy-access to maintenance and service information
12.3 inches (31.2 cm) x 19.3 inches (49 cm) x 7.1 inches (18 cm)
15 pounds, 18.4 pounds with battery
3 Year Manufacturers Warranty on Unit
1 Year Warranty on Accessories
Alert / Alarm:
Pulse Dose Mode, Lower Battery Power, Loss or No Power, Unit Failure, No Inspiration Detected, Low Therapeutic O2 Output, O2 Flow Outside Normal Limits
Back Up Alarm Power:
9 Volt Battery
It may take up to 5 hours, dependent upon your flow setting, to achieve up to 80 percent capacity from a fully empty battery.
Up to 3.7 hours at 1.0 LPM continuous flow
Up to 1.3 hours at 3.0 LPM continuous flow
Up to 4.4 hours at 1.0 pulse flow setting
Up to 2.1 hours at 6.0 pulse flow setting
Pulse Dose: 16 - 96 in 16 mL increments, 128, 160 and 192 mL
Continuous Flow: 0.5 to 3.0 Liters Per Minutes in .5 increments
40 Decibels @ 3.0 Pulse Dose, 48 Decibels @ 3 Liters per Minute on Continuous
Altitude: Sea level to 13,123 feet
Temperature: 50 to 104 degree Fahrenheit @ 82.4 Fahrenheit non-condensing, dew point.
5.0 Nominal PSI (Pounds per Square Inch)
90% @ sea level (+/- 3%)
AC Operation: 100-240 VAC, 50-60 Hz
DC Input: 13.5-15 VDC, 10A
52 watts at 1.0 and 145 watts at 3.0 Continuous Flow
45 watts at 1.0 and 95 watts at 6.0 Pulse Dose
Always keep clean, dry and handle it with care. Store in a safe, accessible area.
Adjustable trigger ranging from -.125 cm to -0.4 cm
How to change filter on the SeQual Eclipse 3?
The SeQual Eclipse 3 has a washable/external filter and an internal filter that need to be tended to. The washable/external filter is located on the back side of the machine just below the carry handle and is a mesh square filter. The gross particle filter should be rinsed off and completely dry before placing back into the machine. Based off the use on the machine hour wise, the internal filter should be replaced every 1-2 years.
How do I perform a hard reset on the SeQual eclipse 3?
To perform a hard reset, you will need to remove the battery from the machine and disconnect all power to the unit. Let the machine stand alone without any power for about 2 minutes. Plug the machine back into AC power (power supply for the house) WITHOUT BATTERY and turn the machine on. While the unit has been running for about 5-10 minutes place the battery into the machine WHILE RUNNING and this should eliminate the issue the machine is having. If the issue still occurs, have the customer call back and then we will follow the process of creating a case.
How do I insert and remove the battery on the SeQual Eclipse 3?
The easiest way to insert and re