SeQual’s Patented autoSAT Technology.
Pulse Dose Mode will deliver a continuous flow of oxygen, always searching for a breath about every 15 seconds.
Approved for Prescription Oxygen in continuous flow mode of 0.5 - 3.0 liters per minute.
LCD screen that shows LPM and mL on the display, as well as easy touch control settings that function on fast, medium and slow settings.
The Universal Cart has a new modern design that allows for trouble-free access to battery. Both AC and DC power adapters are smaller and lighter weight, yet more durable. The AC/DC and battery power choices increase therapy options and allow for more travel opportunities.
FAA approved for most commercial flights. Also approved for use on buses, cruise ships and most other methods of travel.
Wide array of accessories to help support an active lifestyle
Easy access to maintenance and service information
12.3 inches wide x 7.1 inches deep x 19.3 inches high
15 pounds, 18.4 pounds with battery
Three-year Manufacturer’s Warranty on Unit
One-year Warranty on Accessories
Alert / Alarm:
Pulse Dose Mode, Lower Battery Power, Loss or No Power, Unit Failure, No Inspiration Detected, Low Therapeutic O2 Output, O2 Flow Outside Normal Limits
Back Up Alarm Power:
9 Volt Battery
It may take up to 5 hours, depending on the flow setting, to achieve 80 percent capacity from a completely uncharged battery.
Up to 3.7 hours at 1.0 LPM continuous flow
Up to 1.3 hours at 3.0 LPM continuous flow
Up to 4.4 hours at 1.0 pulse flow setting
Up to 2.1 hours at 6.0 pulse flow setting
Pulse Dose: 16 - 96 in 16 mL increments, 128, 160 and 192 mL
Continuous Flow: 0.5 to 3.0 Liters Per Minutes in 0.5 increments
40 Decibels @ 3.0 Pulse Dose, 48 Decibels @ 3 Liters per minute on continuous flow
Altitude: Sea level to 13,123 feet
Temperature: 50 to 104 degree Fahrenheit @ 82.4 Fahrenheit non-condensing, dew point.
5.0 Nominal PSI (Pounds per Square Inch)
90% @ sea level (+/- 3%)
AC Operation: 100-240 VAC, 50-60 Hz
DC Input: 13.5-15 VDC, 10A
52 watts at 1.0 and 145 watts at 3.0 Continuous Flow
45 watts at 1.0 and 95 watts at 6.0 Pulse Dose
Always keep clean, dry and handle it with care. Store in a safe, accessible area.
Adjustable trigger ranging from -.125 cm to -0.4 cm
Items 1 to 10 of 18 total
How to change filter on the SeQual Eclipse 3?
The SeQual Eclipse 3 has a washable/external filter and an internal filter that need to be maintained. The washable/external filter is located on the back side of the machine just below the carry handle and is a mesh square filter. The gross particle filter should be rinsed off and completely dry before placing back into the machine. Depending on the number of hours the machine is used each day, the internal filter should be replaced every 1-2 years.
How do I perform a hard reset on the SeQual eclipse 3?
To perform a hard reset, you will need to first remove the battery from the machine and disconnect all power to the unit. Let the machine sit powered down for 2 minutes. Plug the machine back into AC power (the power supply for use in the home) WITHOUT BATTERIES and turn the machine on. Once the unit has been running for 5-10 minutes, place the battery into the machine WHILE RUNNING. This should eliminate the issue the machine is having. If the issue still occurs, call us and we will begin the process of creating a case.
How do I insert and remove the battery on the SeQual Eclipse 3?
The easiest way to insert and re
This product is available for FREE GROUND SHIPPING*. Please allow 3-6 business days for UPS Ground Delivery in the US; most orders process within 1 business day. This item can be expedited for an additional fee.
Other Delivery Options:
UPS 2nd Day: Delivery the second business day.
UPS Next Day: Delivery the next business day.
Additional charges apply to products shipped to AK, HI and territories outside continental US. * Free Ground Shipping does not apply to AK or HI.