Intuitive, user-friendly screen
Attractive, lightweight and sleek design that weighs around 10 pounds
Pulse Dose and Continuous Flow (up to 2 LPM) options
Easy to remove, rechargeable/replaceable battery which can supply oxygen for up to 7 hours on a pulse dose setting
Convenient carrying case with a detachable accessory bag and handy cart
11.5 inches wide x 6 inches deep x 10 inches high
10 pounds with battery
Three-year Manufacturer’s Warranty
Alert / Alarm:
System technical fault, Wrong battery, Low battery, Depleted battery, High breath rate, Low oxygen purity, No breath, No flow
2 to 3 hours (approximately) from 0% charge to 100%.
Pulse Dose Setting of 2: 3 hours
Continuous Flow of 1 LPM: 1.6 hours
Includes Internal Bacteria Filter
0.5 to 2 liter per minute (LPM) in 0.5 increments
1=12 ml; 1.5=18 ml
2=24 ml; 2.5=30 ml
3=36 ml; 3.5=42 ml
4=48 ml; 4.5=54 ml
5=60 ml; 5.5=66 ml
Around 43 decibels at a setting of 2 (As loud as a refrigerator humming)
41 - 104 Fahrenheit (5 - 40 Celsius)
Up - 10,000 feet (3048 m)
87 to 96% at all Pulse/Continuous Settings
Output 19 VDC 7.9 A max
AC Power Input: 100 to 240 VAC, 50/60 Hz, 2.0 A
DC Power: Input 12 to 19 VDC 8.0 A max
120 watts and 150 watts during charging
-4 - 140 Fahrenheit (-20 - 60 Celsius) unit only
How to change filter on the SimplyGo?
The Simply Go does not have a washable filter, only an internal filter that needs to be replaced. Depending on the number of hours the machine is used each day, the internal filter should be replaced every 1-2 years.
How do I perform a hard reset on the SimplyGo?
To perform a hard reset, you will need to first remove the battery from the machine and disconnect all power to the unit. Let the machine sit powered down for 2 minutes. Plug the machine back into AC power (the power supply for use in the home) WITHOUT BATTERIES and turn the machine on. Once the unit has been running for 5-10 minutes, place the batteries into the machine WHILE RUNNING. This should eliminate the issue the machine is having. If the issue still occurs, call us and we will begin the process of creating a case.
Can Simply Go be used with a CPAP machine?
Yes, the Simply Go unit is continuous flow and will adapt to most CPAP machines.
How do I insert and remove the battery on the SimplyGo?
When looking at the machine from the top where the handle is located, on the left side top corner you will notice a rectangular cutout. This is where the battery is located. When placing the battery inside the machine, make sure you set it in gently. If you drop or slam the battery into the machine, the tabs or pc boards on the battery or the inside of the machine can get damaged or break. When removing the battery, place your fingertips on the top left corner, then just slightly on the backside of the machine where you will notice an indent in the battery. Lift up on the battery and remove toward you, making sure you have a firm grip on the battery so it does not drop back into the machine or slip out of your hand.
How long does the battery last at all settings on the SimplyGo?
The battery duration FULLY CHARGED for Pulse and also Continuous
Setting 1: 3.7 hours
Setting 2: 3.5 hours
Setting 3: 3.1 hours
Setting 4: 2.7 hours
Setting 5: 2.3 hours
Setting 6: 1.9 hours
Setting 1: 2.3 hours
Setting 2: 0.7 hours
This product is available for FREE GROUND SHIPPING*. Please allow 3-6 business days for UPS Ground Delivery in the US; most orders process within 1 business day. This item can be expedited for an additional fee.
Other Delivery Options:
UPS 2nd Day: Delivery the second business day.
UPS Next Day: Delivery the next business day.
Additional charges apply to products shipped to AK, HI and territories outside continental US. * Free Ground Shipping does not apply to AK or HI.