Surprisingly compact design for easy travel and more enhanced portability
Oxygen Sensor that ensures accurate O2 delivery and a reduced maintenance schedule
Less than 20 pounds when the battery installed
Battery life up to 5.4 hours on Pulse that quickly recharges in 2 to 4 hours
Can be used with 35 foot cannula in continuous flow and 50 foot cannula on continuous
12.3 inches wide x 7.1 inches deep x 19.3 inches high
15.5 pounds, 19 pounds with battery
Three-year Manufacturer’s Warranty
Alert / Alarm:
High Flow/Low Flow, No Breathe, Low Oxygen Output, High Temperature, Low Battery, Power Failure, Unit Malfunction
2.0 to 4.5 hours depending on flow setting
3 hours on a setting of 6 and 5 hours on a setting of 1 in pulse dose
1.6 hours on a setting of 3 and 4 hours on a setting of 1 on continuous
Continuous Flow: 1 - 3 LPM
Pulse Dose: 1 - 6 LPM
40 Decibels at 3.0 LPM
Temperature: 41 F to 104 F (5 C to 40 C)
Altitude: 0 to 13,123 ft (0 to 4000 meters)
91% (+/-3%) @ all flow settings
Temperature: -4 F to 140 F (-20 C to 60 C)
Keep dry and clean and handle with care.
Items 1 to 10 of 12 total
How to change filter on the Devilbiss iGo?
The Devilbiss iGo has a washable/external filter and an internal filter that need to be maintained. The washable/external filter is located on the back side of the machine built into the carry handle. If you place your finger deep enough in the carry handle portion there is a mesh rectangle filter. The gross particle filter should be rinsed off and completely dry before placing back into the machine. Depending on the number of hours the machine is used each day, the internal filter should be replaced every 1-2 years.
How do I perform a hard reset on the Devilbiss iGo?
To perform a hard reset, you will need to first remove the battery from the machine and disconnect all power to the unit. Let the machine sit powered down for 2 minutes. Plug the machine back into AC power (the power supply for use in the home) WITHOUT BATTERY and turn the machine on. Once the unit has been running for 5-10 minutes, place the battery into the machine WHILE RUNNING. This should eliminate the issue the machine is having. If the issue still occurs, call us and we will begin the process of creating a case.
How do I insert and remove the battery on the Devilbiss iGo?
The easiest way to insert and remove the battery is to have the machine standing upright. There is a small push down tab at the top of the machine. Press down on this tab, and the battery clips away from the machine. When placing the battery inside the machine, make sure to set it in gently. If you drop or slam the battery into the machine, the tabs or pc boards on the battery or the inside of the machine can be damaged or break. When removing the battery, you must press down on the battery tab then pull the battery away from the unit. Make sure you have a firm grip on the battery so it does not drop back into the machine or slip out of your hand.
How long does the battery last at all settings on the Devilbiss iGo?
The battery duration FULLY CHARGED for Pulse and also Continuous. Keep in mind that battery durations are estimated because each patient has different breath rates and breathing patterns, which will make the batteries last for shorter or longer amounts of time. When you increase or decrease the settings, the charge per battery will either increase or decrease.
Setting 1: 5.4 hours
Setting 2: 4.7 hours
Setting 3: 4.0 hours
Setting 4: 3.5 hours
Setting 5: 3.2 hours
Setting 6: 3.0 hours
Setting 1: 4.0 hours
Setting 2: 2.4 hours
Setting 3: 1.6 hours
This product is available for FREE GROUND SHIPPING*. Please allow 3-6 business days for UPS Ground Delivery in the US; most orders process within 1 business day. This item can be expedited for an additional fee.
Other Delivery Options:
UPS 2nd Day: Delivery the second business day.
UPS Next Day: Delivery the next business day.
Additional charges apply to products shipped to AK, HI and territories outside continental US. * Free Ground Shipping does not apply to AK or HI.